- Garden / Landscaping / Patio — 208
- Kitchen / Bathrooms — 208
- Appliance / Repair — 158
- Interior Design / Decor — 145
- Real Estate / Finance — 119
- Floors / Tile / Hardwood — 105
- Doors / Garages — 97
- HVAC / Air Conditioning — 92
- Cleaning / Maintenance — 87
- Construction / Materials — 82
- Plumbing / Basements — 80
- Safety / Security — 77
Residual Chlorine Analyzer Applications for Pools
by Guest on Jun 6, 2011
Chlorine meters are used in many applications requiring the measurement of residual chlorine concentration for chlorine monitoring or residual chlorine dosing control.
A residual chlorine analyzer or meter has various applications including:
- Continuous chlorine monitoring
- Water treatment plant residual chlorine control
- Secondary chlorination free chlorine control
- Distribution monitoring for total residual chlorine
- Cooling tower chlorine monitoring and control
- Pasteuriser chlorine dosing control
- Seawater chlorination control
- Chlorine in seawater monitor
The membraned amperometric chlorine sensor is enhanced with a third, reference, electrode which eliminates zero drift. pH correction is often not required at all. (NB. These chlorine sensors are often known as polarographic chlorine sensors although this is a misuse of the word polarographic).
The chlorine sensor used by the Residual Total Chlorine Monitors and the Residual Free Chlorine Monitors are both largely pH independent meaning that both chlorine analyzers are bufferless and reagentless. They both utilize amperometric chlorine sensors and both show remarkable sensitivity and stability. For those making a chlorine measurement at high pH (>pH 8.5) on variable pH water it is possible to provide pH compensation from either a pH sensor connected to the chlorine analyser or from an external pH meter.
The chlorine sensors work by separating the electrodes that perform the chlorine measurement from the sample, by a membrane. This membrane allows the free residual chlorine (HOCl and OCl-) or the Total Residual Chlorine (HOCl and OCl- plus chloramines) through the membrane. Inside the sensor the residual chlorine meets the electrolyte which is at a low pH. This converts the majority of the OCl- to HOCl. The HOCl is reduced at the gold working electrode and the current generated is proportional to the Free Chlorine or Total Chlorine present, and the chlorine analyzer gives a reading in chlorine ppm or mg/l.
This technique is the most advanced method of continuous chlorine measurement and has many benefits to the user including a very stable chlorine measurement and better chlorine dosing control.
In addition to residual chlorine measurement the free chlorine sensor determines the chlorine concentration and the chlorine analyzer can control the chlorine dosing using a variety of control algorithms including PID control.
The Residual Chlorine Monitor is bufferless and reagent free, meaning it has a low total cost of ownership and with maintenance intervals at 3 or even 6 months, the HaloSense bufferless Chlorine Monitor is fast becoming the instrument of choice for the engineer who wants the best chlorine monitor at the best price.
The HaloSense Free and Total Residual Chlorine Analyzers are equipped with process control options, datalogging, relay outputs, serial comms, (TCP, Ethernet, ModBus are available now with Profibus available soon). Remote Access of the chlorine monitors (including remote access to all control options) is available via the internet, and via a LAN. In fact the HaloSense range of chlorine analyzer products has all the options you could want in a Chlorine Analyzer whilst maintaining a low cost and great value for money.
Most Recent Articles
- Mar 11, 2019 How Inclement Weather Can Affect Your Pool by Tammy Emineth
- Nov 11, 2018 7 Luxury Pool Ideas For Your Home With Liquid Limestone by Valerie Muscat
- Jun 21, 2018 Should I Try to Tackle Resurfacing My Pool on My Own? by Brenda Vollman
- Oct 31, 2017 Your Quick Guide to Patching up a Vinyl Swimming Pool Liner by Frank Myrland
- Oct 31, 2017 Cheat Sheet for Spa Experience At Your Home by Lori Longoria